View and Download HP A manual change online. Impedance/Gain-Phase. A Multimeter pdf manual download. Hp A Pdf User Manuals. View online or download Hp A Manual Change. Current Directory: /HP Agilent/HP A Impedance Gain Phase Analyzer. Click on a File to download or a Directory to open. File Type. File Size. ===> Back.
|Published (Last):||26 June 2006|
|PDF File Size:||17.70 Mb|
|ePub File Size:||1.96 Mb|
|Price:||Free* [*Free Regsitration Required]|
I no longer have acces to a A manual, but Agilent had a number of tech notes for the A, and impedance measurements in general, on their website. I had a quick question on making impedance measurements of a device using the HP A when there is a function generator connected in series with the device.
I am not sure that there is a way where you can separate the interference of the sine function from the Imp analyzer and the constant 2MHz from your FG. The device acts as a resistor for frequencies greater than kHz and when the AC field is turned off it loses its resistive behavior. Dec 248: The free PDF on Agilent’s web site has unreadable schematics. The HP A also features high accuracy and error elimination functions to ensure reliable measurements. By using measured values, this unique function can approximate the circuit constant values of five circuit models.
Heat sinks, Part 2: Most Active Hardware Boards: Distorted Sine output from Transformer 8.
The analysis function not only provides you with impedance and transmission characteristics, but also allows you to determine secondary parameters.
Message 18 of Using HP A’s Equivalent Circuit Analysis Function, you can easily and quickly obtain those equivalent circuit constants that, until now, required a number of time-consuming, 419a4 calculations. Added after 11 minutes: Equating complex number interms of the other 6.
Detailed impedance and transmission characteristics, including secondary parameter derivations, can be simply and quickly evaluated or tested. Manaul accuracy and wide range- Impedance measurement: You can perform the compensations with only your function generator in the circuit, and that should help. Message 11 of Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
The HP A can contribute to improving engineering productivity and reducing test cost. By using ASP and actual measurement values, you can readily calculate many secondary parameters that you may need to evaluate. Your 41944a on this would be greatly appreciated.
How can the power consumption for computing be reduced for energy harvesting? ModelSim – How to force a struct type written in SystemVerilog? The HP A’s high degree of measurement accuracy – 0.
Service manual for an HP A Impedance/Gain-Phase Analyzer
The problem is that we have to manipulate the circuit element with an external AC field via a function generator and simultaneously measure the circuit properties of the active nanostructure as a function of frequency using the impedance analyzer.
Equivalent Circuit Analysis Function Using HP A’s Equivalent Circuit Analysis Function, you can easily and quickly obtain those equivalent circuit constants that, until now, required a number of time-consuming, complicated calculations. For example, a resonator’s equivalent circuit elements or a coil’s self inductance, lead resistance, and stray capacitance can be easily obtained. The frequency sweep done by the HP analyzer does not overlap the AC frequency applied by the function generator.
HP 4194A Manual Change
Part and Inventory Search. Hierarchical block is unconnected 3. Realistic device characteristics can be analyzed mannual actual 419a4 conditions by varying the test frequency, test signal level, and dc bias.
Service Manual for Tek Spectrum Analyzer 0. Message 19 of Message 17 of View Cart My Account. I think that the analyzer is essentially a function 41994a. The HP A’s ASP function, an internal programming feature, allows you to control all HP A operations measurement, display, and analysis without the need for an external computer. I chose this frequency range because the device behaves like a resistor in this frequency range. The device I am testing is a planar gold electrode system.
If I actuated from say Hz – 1 kHz using the analyzer, it would change the properties of the device.